|The cookie settings on this website are adjusted to allow all cookies so that you have the very best experience. If you continue without changing your cookie settings, we'll assume that you are happy to receive all cookies on our website. However, if you would like to, you can change your settings at any time using the Change cookie settings link in the Special menu.|
Transit Time Ultrasonic Flow Meters
Dynasonics offers a full line of transit time ultrasonic flow meters to satisfy a variety of customer applications. The TFX family of products features non-invasive meters designed to provide superb accuracy and response characteristics for measuring clean liquids.
Transit time flow meters utilize two transducers which function as both ultrasonic transmitters and receivers. The flow meters operate by alternately transmitting and receiving a frequency modulated burst of sound energy between the two transducers. The burst is first transmitted in the direction of fluid flow and then against fluid flow. Since sound energy in a moving liquid is carried faster when it travels in the direction of fluid flow (downstream) than it does when it travels against fluid flow (upstream), a differential in the times of flight will occur. The sound's time of flight is accurately measured in both directions and the difference in time of flight calculated. The liquid velocity (V) inside the pipe can be related to the difference in time of flight (dt) through the following equation: V = K * D * dt, where K is a constant and D is the distance between the transducers